For each use, set the instrument's switch to the "ON" position with the "ON" switch on and the LCD shows "0" mj / cm2 reading (millijoules per square centimeter). For non-specific use, Before a measurement, the readings to zero.
If your process special needs, but also can be measured repeatedly, after each measurement readings, do not need to zero processing, then the instrument last displayed readings will be repeated measurements of the cumulative value.
When the measurement is completed, the switch must be set to the "OFF" position so that the instrument can be shut down in time. The correct operation will prolong the battery life of the instrument.
- The Correct Operation Will Effectively Extend T...
- The Nature Of The Power Can Be Divided Into DC ...
- Effective Correction Of Time Difference And Degree
- With Buffer Storage And Intelligence
- Cost-effectiveness, Intelligence And Reliability
- Efficient, Energy Saving, Safety And Environmen...
- Increased Complexity And Reduced Network Capacity
- Reduce Customer Waiting Time And Improve User E...
- The Real Purpose Of Saving Energy
- Improve The Relationship Between Man And Nature
- Energy Meter Use Precautions
- Thermal Detector With Introduction
- Thermal Detector With Introduction Fundamental
- Thermal Detector With Types Of
- Thermal Detector With Characteristic
- How Does Energy Meters Work
- Household Electricity Saving Tips
- How Does A Single Electronic Prepaid Electric E...
- Smart Wattmeter Parameters And Applications